Flow Probe uses true velocity averaging. To start a running average,
first hold the reset button for 3 seconds to zero the display. When the
probe is placed in the water flow, an average will begin. As long as
thewater velocity sensor remains in the flow, the averaging continues.
Once the average reading becomes steady, the true average velocity has
been obtained. When you remove the Flow Probe from the water, the
average value is frozen on the display until reset.Flow Probe Specifications :
Range: 0.3-19.9 FPS (0.1-6.1 MPS)
Accuracy: 0.1 FPS
Averaging: True digital running average. Updated once per second.
Display: LCD, Glare and UV Protected
Control: 4 button
Datalogger: 30 sets, MIN, MAX, and AVG
Features: Timer, Low battery warning
Sensor Type: Protected Turbo-Prop propeller with magnetic pickup.
Instrument: 2 lbs. (0.9 kg) (FP111), 3 lbs. (1.4 kg) (FP211), 2.8 lbs. (1.3 kg) (FP311)
Shipping: 13 lbs. (5.9 kg) (FP111), 23 lbs. (10.4 kg) ((FP211), 19 lbs. (8.6 kg) ((FP311)
Expandable Length: 3.7 to 6 ft (1.1 to 1.8 m) (FP111); 5.5 to 15 ft (1.7 to 4.6 m) (FP211); 2.5 to 5.5 ft (0.76 to 1.7 m) (FP311)
Probe: PVC and anodized aluminum with stainless steel water bearing
Computer: ABS/Polycarbonate housing with polyester overlay
Power: Internal Lithium Battery, Approx 5 year life with typical use, Non-Replaceable
Operating Temperature: -4° to 158° F (-20° to 70° C)
Storage Temperature: -22° to 176° F (-30° to 80° C)
Carrying Case: The Flow Probe is shipped in a padded carrying case.