081362449440 Jual LabScan® XE
PD. BINTANG INSTRUMENT
laboratorium, lingkungan, Lab tester
For best performance, your instrument should be placed where there is ample work space with medium
or subdued illumination and no drafts. For optimum results, a clean, air-conditioned area is
recommended with a relative noncondensing humidity of 20-80% [ 10-90% for sensor only] and
relatively constant temperature not exceeding 90° F ( 32° C) [ 104° F ( 40° C) for sensor only] . For
specification performance, the recommended temperature range is 70-82° F ( 21-28° C) .
The instrument should be connected to a stable, instrument-grade power line. If other equipment is
connected to the same power line, a transient power surge may be produced when the other equipment is
turned on. If this happens, restandardize the instrument before making measurements. HunterLab
recommends using a line conditioner with a minimum 600 VA rating and a battery back-up system.
Length - 14.5 inches ( 36.9 cm)
Width - 7.5 inches ( 19.1 cm)
Height - 12.0 inches ( 30.5 cm)
Weight - 23 lbs ( 10.4 kg)
Computer system - depends upon system selected. Refer to manufacturer' s documentation for
Operating temperature: 10 - 40° C ( 50 - 104° F)
Storage temperature: -21 - 66° C ( -5 - 150° F)
Operating humidity ( relative, non-condensing) : 10 - 90%
Voltage: 100-240 VAC, 47/ 63 Hz. * ( See note on page 22-19.)
180 VA maximum.
Fuse: 1.4A, SB.
Installation Category ( Over Voltage) : II.
04/ 02 LabScan XE 22-17
Power Cord Wire Color Code:
220V Cord Brown LINE
Green/ Yellow SAFETY
110V Cord Black LINE
Instrument-Computer Ground Potential Check:
Perform this check if the power cord wiring is being changed, such as for replacement of the cord or if
changing the plug.
The instrument serial port ground connection is referenced to the instrument' s internal frame and safety
ground. Before connecting the communication cable to the instrument serial port, apply power to the
instrument and the host computer. Check the ground potential ( voltage) between the serial port ground
pins on the computer and the instrument. The ground connection is pin 5 on DB-9 connectors and pin 7
on DB-25 connectors. Voltages in excess of 5VAC at 110V input power, or in excess of 10VAC at
220V input power may indicate a difference in ground wiring and can damage the instrument and/ or the
computer. Check the wiring and take other steps as needed to reduce this difference before connecting
the communications cable. You may also use a data optoisolator.